An unusual mode of failure of a tripolar constrained acetabular liner: a case report.
AffiliationAdelaide and Meath Hospital, Incorporating the National Children's Hospital,, Tallaght, Dublin, Ireland. Louisabanks@hotmail.com
Arthroplasty, Replacement, Hip/*adverse effects
Hip Prosthesis/*adverse effects
MetadataShow full item record
CitationArch Orthop Trauma Surg. 2010 Apr;130(4):503-5. Epub 2009 Jul 24.
JournalArchives of orthopaedic and trauma surgery
AbstractDislocation after primary total hip arthroplasty (THA) is the most commonly encountered complication and is unpleasant for both the patient and the surgeon. Constrained acetabular components can be used to treat or prevent instability after primary total hip arthroplasty. We present the case of a 42-year-old female with a BMI of 41. At 18 months post-primary THA the patient underwent further revision hip surgery after numerous (more than 20) dislocations. She had a tripolar Trident acetabular cup (Stryker-Howmedica-Osteonics, Rutherford, New Jersey) inserted. Shortly afterwards the unusual mode of failure of the constrained acetabular liner was noted from radiographs in that the inner liner had dissociated from the outer. The reinforcing ring remained intact and in place. We believe that the patient's weight, combined with poor abductor musculature caused excessive demand on the device leading to failure at this interface when the patient flexed forward. Constrained acetabular components are useful implants to treat instability but have been shown to have up to 42% long-term failure rates with problems such as dissociated inserts, dissociated constraining rings and dissociated femoral rings being sited. Sometimes they may be the only option left in difficult cases such as illustrated here, but still unfortunately have the capacity to fail in unusual ways.
- Failure of constrained acetabular liner without metal ring disruption.
- Authors: Sathappan SS, Ginat D, Teicher M, Di Cesare PE
- Issue date: 2008 Mar
- Salvaging unstable or recurrent dislocating total hip arthroplasty with the constrained acetabular component.
- Authors: Stanton DA, Bruce WJ, Goldberg JA, Walsh W
- Issue date: 2002 Dec
- The use of a constrained acetabular component for recurrent dislocation.
- Authors: Shapiro GS, Weiland DE, Markel DC, Padgett DE, Sculco TP, Pellicci PM
- Issue date: 2003 Apr
- Outcome of the Stryker® Trident 'All-Poly' constraint acetabular insert: a district general hospital experience.
- Authors: Harrison SJ, Leeder DJ, McWilliams TG, Metcalf RW, Sidhom SA
- Issue date: 2015 Nov-Dec
- Failure mechanism of a constrained liner: a case report.
- Authors: Niimi R, Sudo A, Hasegawa M, Uchida A
- Issue date: 2012 Feb